MA-3 Solo Sample Fork Replacemenmt
Remove Sample Fork Assembly from the MA-3 Solo by removing the two screws attached to the sliding rail.
Remove the fork portion from the rest of the assembly by removing the two screws in the center of the assembly.
Attach the replacement fork to the rest of the assembly
Please ensure that the fork is centered (see examples below)
Bad Example Good Example
Reattach the assembly back onto the sliding rail and check for closing functionality.
Sample Fork Replacement
① Turn the power OFF, open the top cover panel, and open the front cover panel. ② With the covers open, power on the instrument and start up the MA-3000 software, and open the maintenance screen. ③ Remove the Teflon joint on the front side of ...
Sample Fork Cleaning
**Please follow the Sample Fork Replacement Procedure to remove the sample fork.** Once the sample fork has been removed you may flush out the sample fork using squirt bottles. As a general rule, flush it first with a steady stream of IPA for at ...
Cleaning Sample Boats
The primary means of cleaning your boats should be by heat treatment. Carry out the heat treatment of a sample boat according to (1) or (2) described below. (1) Repeatedly run empty boats using the MA-3 analyzer until the blank value has been ...
MA-3 Solo Combustion Tube Replacement
The attached PDF and video are the instructions for removing and replacing the combustion tube and gold trap on the MA-3 Solo Instrument.
MA-3 Solo Troubleshooting Chart
Problem Cause Solution There is leak Teflon joints are loose Tighten joints *Run leak check from Maintenance Sample heating tube is broken Replace sample heating tube Inlet O-ring is broken or damaged replace with new O-ring Flowcell has an ...