PE-1000 H5 Heater Thermocouple Replacement
The attached PDF will explain the procedure to replace the thermocouple on the H5 heater on the PE-1000.
MA-3000 H2 Heater Replacement Instructions
MA-3000 H2 Heater Replacement Instructions These instructions are for units with Serial# 12740141 and greater, please check your serial number to confirm these are the correct instructions Inside of the MA-3000 is at a very high temperature. Please ...
MA3000 H3 Heater and Gold Trap Positioning
The attached PDF article will demonstrate the proper placement of the gold trap within the H3 heater coil as well as demonstrate the proper placement of the thermocouple on the heater coil. Please note the thermocouple is wrapped around the eighth ...
PE-1000 Lamp Replacement
The attached PDF may be used to replace, or exchange, the lamp in the PE-1000 instrument. This procedure was taken directly from the manual. You can find the manual here.
PE-1000 Troubleshooting Chart
Problem Cause Solution There is leak T-joint is loose Tighten t-joint Other joints are loose Tighten joints Sample heating tube is broken Replace sample heating tube Sample inlet nut is loose Tighten sample inlet nut Septum is faulty ...
PE-1000 H1 Heater Replacement
First, you will need to follow the procedure to remove the combustion tube. (Instructions Here) Next, remove the nuts circled in Red below and separate the heater from the instrument. Replace the new heater, paying special attention to the ...