NIC MA-3000: Remove the Z-axis Nut
Instructions to Remove the Z-axis Nut on the NIC MA-3000 Mercury
- Remove the two screws seen in the first photo
- This will free up the plate that holds the Z-Axis (bottom photo)
- Move the plate out of the way and remove the bearing on the end
(bearing pulls off)
- Remove the 4 screws that hold the z-axis nut in place (2nd photo)
- Remove the nut
Replacing the Autosampler Arm Assembly - NIC RA-4 Series
This operation should only be performed after consultation with AGS or NIC service personnel. 1) With the RA-4 powered ON, click the top right icon in the software to move the arm to the down position. 2) Power OFF the analyzer. 3) Remove the ...
MA-3000 Autosampler Lift Adjustment
This procedure should only be performed after speaking with a service technician 1. Remove the entire Tray Holder assembly by removing the 4 screws in the corners 2. Adjust the motor position by centering the shaft by loosening two screws (yellow ...
SC-3 Autosampler Alignment
Pre-Checks 1. Looking down on the autosampler head, there are two screws that go through metal springs and attach to the cap. Ensure that they are snug, but do not strip them. 2. Run a few tests sample runs to see if there is a consistent pattern ...
MA-3000 Autosampler Adjustment
Open Maintenance Tab and select "BC" Tab; then select BC Adjust. The Password is 0219 Place an empty boat in one of the positions in a rack. You will now follow the on-screen instructions to make the necessary adjustments to the sampler. You will ...
MA-3000 Price List with Pictures
The attached PDF file contains the part numbers, prices and frequency of replacement for the most common parts for this mercury analyzer. To place an order, please send an email to email@example.com or call us at (979) 320-0052.