MA-3 Solo Manual Leak Check
This manual leak check procedure should be run if the auto leak check fails during self test to help identify the location of the leak
If bubbling does not occur, or is weak, in the humidification bottle proceed with the following steps:
MA-3 Solo Sample Fork Replacemenmt
Remove Sample Fork Assembly from the MA-3 Solo by removing the two screws attached to the sliding rail. Remove the fork portion from the rest of the assembly by removing the two screws in the center of the assembly. Attach the replacement fork to ...
MA-3 Solo Troubleshooting Chart
Problem Cause Solution There is leak Teflon joints are loose Tighten joints *Run leak check from Maintenance Sample heating tube is broken Replace sample heating tube Inlet O-ring is broken or damaged replace with new O-ring Flowcell has an ...
(1) Run a leak check from the maintenance screen and follow the onscreen instructions. If it fails the leak check, first ensure that the cap is correctly aligned on the top of the sample tube. If it is not, follow this procedure before continuing. ...
MA-3 Solo Detector Flowcell Removal and Cleaning Instructions
The attached PDF are the instructions to remove, disassemble, and clean the detector flowcell of the MA-3 Solo instrument.
MA-3000 User Manual
The attached PDF is a copy of the MA-3000 User instrument and operational manual. Most, if not all, of the maintenance procedures can be found in this manual.