MA-3 Solo Heater Failure Troubleshooting
MA-3 Solo Troubleshooting Chart
Problem Cause Solution There is leak Teflon joints are loose Tighten joints *Run leak check from Maintenance Sample heating tube is broken Replace sample heating tube Inlet O-ring is broken or damaged replace with new O-ring Flowcell has an ...
MA-3000 Troubleshooting Chart
Problem Cause Solution There is leak Joints are loose Re-tighten all joints; fitting on T-joint and trap may need 1/4 turn with small wrench Most Common Sources of Leaks Sample heating tube is broken Remove and Install new sample heating tube ...
Sediment Application Note -MA-3000/ MA-3 Solo
The attached PDF is an application note for running sediment on a MA-3000 and MA-3 Solo.
MA-3 Solo Sample Fork Replacemenmt
Remove Sample Fork Assembly from the MA-3 Solo by removing the two screws attached to the sliding rail. Remove the fork portion from the rest of the assembly by removing the two screws in the center of the assembly. Attach the replacement fork to ...
MA-3 Solo Detector Flowcell Removal and Cleaning Instructions
The attached PDF are the instructions to remove, disassemble, and clean the detector flowcell of the MA-3 Solo instrument.