MA-3000 Autosampler Lift Adjustment
This procedure should only be performed after speaking with a service technician
Remove the entire Tray Holder assembly by removing the 4 screws in the corners
2. Adjust the motor position by centering the shaft by loosening two screws (yellow circle)
3. Re-adjust the motor drive (red circle) until the lift is moving smoothly
This adjustment is very sensitive and may need to be performed several times until it is moving smoothly
MA-3000 Autosampler Adjustment
Open Maintenance Tab and select "BC" Tab; then select BC Adjust. The Password is 0219 Place an empty boat in one of the positions in a rack. You will now follow the on-screen instructions to make the necessary adjustments to the sampler. You will ...
RA-4300 Autosampler Arm Adjustment
This procedure can be used to adjust the arm so that it is properly sealing on the sample test tubes RA-4300 Autosampler Adjustment 1. Power on the RA-4300 and open the software 2. Open Run→Maintenance 3. Place a blank tube in position 1, 47, and 77 ...
PE-1000 Autosampler Turret adjustment
The attached PDF is a document that can be used to adjust the position of the turret on the autosampler.
SC-3 Autosampler Alignment
Pre-Checks 1. Looking down on the autosampler head, there are two screws that go through metal springs and attach to the cap. Ensure that they are snug, but do not strip them. 2. Run a few tests sample runs to see if there is a consistent pattern ...
RA-4 Autosampler Height Adjustment
This procedure can be used to adjust the autosampler height to accomodate sample tubes that have a different height than the ones that came with the Instrument 1. Measure the sample tube length 2. Calculate difference between the sample tube you are ...